ASTM D – Standard Test Method for Dielectric Breakdown Voltage of Insulating Oils of Petroleum Origin. Using VDE Electrodes. ▫ IEC Insulating. Page 1. Page 2. Page 3. Page 4. Find the most up-to-date version of ASTM D at Engineering
|Published (Last):||13 February 2010|
|PDF File Size:||6.22 Mb|
|ePub File Size:||3.37 Mb|
|Price:||Free* [*Free Regsitration Required]|
Do not use a low boiling point solvent, as its rapid evaporation may cool the cell, causing moisture condensation. Work Item s – proposed revisions of this standard. A cell having a capacity of 0.
Phenix Technologies TC/VDE-100 Test Cell (LD100) for ASTM D1816 (with stirrer)
Calibrate and label automatic controls in terms of rate-of-rise. However, a high dielectric breakdown voltage does not astj indicate the absence of all contaminants; it may merely indicate that the concentrations of contaminants that are present in the liquid between the electrodes are not large enough to deleteriously affect the average breakdown voltage of the liquid when tested adtm this test method see Appendix X1.
Email Us or call v1816 At the beginning of each days testing, clean according to 7. The voltage in the test circuit should be stable enough to be unaffected by varying current flowing in the capacitive and resistive paths of the test circuit. The impeller, located below the lower edge of the electrodes, rotates in such a direction that the resulting liquid flow is directed downward against the bottom of the test cell.
Oil sampling procedures are detailed in Practice D The residue from the buffing should be removed by repeated wiping with lint-free tissue paper saturated with a suitable solvent, followed by solvent rinsing or ultrasonic cleaning.
Pharmaceutical laboratory apparatus stability chambers, tablet disintegration, tablet dissolution, tablet hardness, V-blenders, If the test of any sample is below the breakdown value being used by the operator as a minimum satisfactory value, drain the cell and flush the cell with new oil of the type to be tested that is filtered through a 5-micron filter and containing less than 25 ppm moisture before testing the next specimen.
Up to 75kV rms symmetrical Voltage measurement accuracy: The spacing agreed upon shall be measured with the gage that corresponds exactly to the selected spacing within tolerance stated above for the gage.
The quality of the insulating oil must be checked at regular intervals to ensure a long equipment service life. Contact with us now Your mail: Right-click to wstm the PDF in a new window or download it: It is important to avoid touching the electrodes or the cleaned gage with the fingers or with portions of the tissue paper or chamois that have been in contact zstm the hands.
Petroleum Products and Biofuel lab apparatus baths, calorimeters, color measurement, density meters, flash point, hydrogenators, oil testing, penetrometers, surface tension, vapor pressure, viscometers and rheometers, Determination of the dielectric breakdown voltage of insulating oils provides an early detection method for any reduction in the insulating properties.
Construct the test cell of a material of high dielectric strength, that is not soluble in or attacked by any of the cleaning or test liquids used, and is nonabsorbent to moisture and the cleaning and test liquids.
Clearance from the electrodes to all sides, bottom, cover or baffle, and any part of the stirring device is at least It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Enter product namemake or model. When not in d186, keep the cell filled with oil that meets the requirements of Specification Asmt of the type normally tested.
Koehler K 75kV Dielectric Strength Tester (ASTM D, D) | Labequip
Particular reference should be made to the general precaution statement of this test method. What are your requirements. Referenced Documents purchase separately The documents listed below are referenced within the subject standard but are not provided as part of the standard. If it is necessary to readjust the electrodes, lock the electrodes and check the spacing. Looking for a specific close-up?
Hipotronics TCVDE Oil Dielectric Test Cell for ASTM D testing – Mitchell Instrument Company
Plastic bottles are permeable and moisture content of the sample may change resulting in a measurable difference when compared to samples collected in non-permeable containers.
This test method should be used as recommended by professional organization standards such as IEEE C The test cell shall be designed to permit easy removal of the electrodes for cleaning and polishing, verification that the shape is within the specified tolerance, and to permit easy adjustment of the gap spacing.
The sample shall be taken in a dry, clean, non-permeable bottle. Measurement of oil temperature: Adjustment and Care of Electrodes and Test Cell 7. The dielectric breakdown voltage serves to indicate the presence of contaminating agents such as water, dirt, cellulosic fibers, or conducting particles in the liquid, one or more of which may be present in significant concentrations when low breakdown voltages are obtained.
If this occurs, before using, warm the cell to evaporate the moisture. Provide the test cell with a motor-driven two-bladed impeller and drive shaft, constructed of a material having high dielectric strength.
General Use laboratory d816 balances, baths, blenders, centrifuges, chart recorders, freezers, fume hoods, furnaces, glassware washers, glove boxes, mixers, moisture testing, ovens, pumps, shakers, vacuum equipment, water purification After thorough cleaning, flush the cell with new oil of the type to be tested that axtm filtered through a 5-micron filter and containing less than 25 ppm moisture. Power by seonoco All Rights Reserved: Historical Version s – view previous versions of standard.
Avoid touching the electrodes or the inside of the cell after cleaning. It has been found to be especially useful in diagnostic and laboratory investigations of the dielectric breakdown strength of insulating liquid in insulating systems.